This single-width module uses 16 bits of data. These bits are loaded into one of the three register counters. At the end of the write command the register counts down until the counter reaches zero. The counter may be started also from an external negative-going signal on the I/O connector. The train of pulse at 100 KHz rate is equal to the number of binary bits set.
|Reads module number
|Load register 1 and start train of pulses from register 1
|Load register 2 and start train of pulses from register 2
|Load register 3 and start train of pulses from register 3
|Clears all registers
The following lights and switches are provided on the front panel
|Flashed whenever module is addressed
|Flashes when register one is outputting pulses
|Flashes when register two is outputting pulses
|Flashes when register three is outputting pulses
All BNCs are transformer-coupled.
Output is 1uS pulse width, 10V into 50 Ohm at 100Khz
|Register 1 output
|Register 2 output
|Register 3 output